Artifact from Frayed Leads Obscures STEMI
Artifactual activity on 12-lead EKG presents a significant impediment to electrocardiographic diagnosis. A case is presented here in which underlying STEMI could not be appreciated due to artifactual interference from frayed electrode leads. Clinicians should to be aware of the causes and presentations of EKG artifact in order to avoid similar pitfalls.
An “all fields” PubMed search was conducted using the term “artifact” in conjunction with each of the terms “STEMI”, “myocardial infarction”, “EKG”, “ECG”, and “ST segment.” Results yielded 0, 76, 19, 317, and 22 references respectively. These 434 citations were then screened for relevance according to title. The scope of the search spanned from 1973 to June 2012.
Numerous sources and types of artifactual interference on EKG have been identified. Artifact may be defined as any electrical activity present on EKG recording which does not directly and appropriately reflect cardiac activity. Artifactual interference may be classified as either of primary, non-cardiac etiology, or of secondary etiology when authentic cardiac signals are deranged due to incompetent acquisition, processing, or presentation. In the former category, a multitude of electrical and mechanical devices have been implicated (1-12, 46). Movement artifacts such as patient tremor, respiration, coughing, and hiccups have also been described (13-21, 43, 44). Artifact resulting from bed or stretcher movement should also be included in this subgroup.
Regarding the derangement of authentic cardiac signals rather than non-cardiac interference, investigators have noted an extensive variety of effects due to electrode misplacement (25-28, 32, 37). Acquisition filters have also been found to deceptively alter the appearance of the electrocardiogram (30, 33). Inconsistent electrode contacts as well as flawed or inverted lead connections can be problematic (45). Printers, monitors, and electronic transmission software have all been implicated in significant distortion or augmentation of the EKG (29, 41).
Too numerous to count case reports involving both primary non-cardiac interference as well as secondary artifact effects have illustrated a diversity of arrhythmic, ischemic, and other electrocardiographic mimics. Typically low frequency primary artifact resulting from tremors or rhythmic movement of physiologic cycle length has been associated with the mimicry of dysrhymias, often wide complex dysrhthmias (13-21, 23). Derangements of authentic cardiac activity resulting from lead reversals, filtering effects, and post-acquisition processing have frequently been associated with the mimicry of ischemic EKG patterns. The appearance of pathologic Q-waves, dramatic changes in cardiac axis, T-wave deflection, and alterations of R-wave amplitude and progression have been documented (25-27). False ST elevation and depression have also been described (30, 31). Both the masking of intrinsic pathology and the pathologic representation of healthy cardiac signals have been noted (30-34, 40, 45). The consequences of unrecognized artifactual interference can include inappropriate pharmacological and electrical therapies; significant morbidity and mortality has resulted (15, 18, 19, 28).
In some cases, the clinician can exploit artifactual activity. Shivering artifact in the presence of electrocardiographic evidence of hypothermia is such a case (42). The utility of respiratory artifact has also been explored (24). More recently, the exploitation of systematic computer algorithm interpretation error has been discussed relative to “double counting” of heart rate in the setting of hyperkalemia (41).
In this case report, an anterior ST-elevation myocardial infarction was masked by opaque artifactual activity resulting from frayed electrode leads. To date, this would appear to be the first documented case of such an occurrence.
An 85 year-old Caucasian man with a history of atrial fibrillation and anxiety awoke at 2:30 AM with chest pressure and shortness of breath. He alerted his daughter and she administered his Xanex, believing his symptoms to be psychosomatic. When this had little effect, an ambulance was called. On their arrival at 3:40 AM, paramedics administered oxygen and 162mg of aspirin. Vital signs at this time were within normal limits. A rhythm strip was acquired which demonstrated heavy artifact obscuring all but one lead. Additional leads were not visualized and no intelligible 12-lead could be obtained.
The patient was transported to a non-PCI capable community hospital. There, a 12-lead EKG was recorded which showed explicit anterior wall STEMI.
The troponin was 0.65. Tenectaplase was administered at 4:30AM; a repeat EKG 90 minutes later was unchanged. At this time, he was transferred to an outside hospital for cardiac catheterization.
On arrival at 7:05AM, the patient was hypotensive with a systolic blood pressure of 70mmHg.
An aortic balloon pump was placed and dopamine initiated. A complete occlusion of the mid-LAD was identified; thrombectomy was performed and the vessel stented with TIMI3 result. Hypotension persisted and the patient developed increasing lethargy and dyspnea. He vomited and became apenic while in cath lab. At 8:15AM he was intubated and placed on levophed; his ejection fraction was less than <15%. Hypotension remained refractory despite the addition of vasopressin and dobutamine. At 10:25AM, the troponin was 92. At this time he was unresponsive on exam with central cyanosis and mottling to all four extremities. There was pulmonary edema with an arterial line indicating a systolic BP of 50mmHg. Blood gas analysis indicated a pH of 7.10. He was described as not likely to survive and made DNR at 10:50AM. At 11:58 AM no carotid pulse could be appreciated and he was pronounced dead.
Retrospective analysis of the prehospital EKG artifact was undertaken. The system was traced from the electrode-lead junctions back to the monitor. In this case, a Physio-Control Life Pack 12 device was being utilized and revealed cable-junction fraying. Experienced operators of this device are often familiar with this type of artifact, and the cable-junction is a known weak point.
Cable fraying or, more broadly, lead-connection artifact, has a distinct electrocardiographic signature. Fequentlely there is an erraticly wandering baseline with sharp, irregular voltage spikes showing inconsistantly varrying amplitudes. As usualy only one connection is effected, the artifact should localize to a particular lead. Thus there should also be leads present which are free of artifact.
Note that in the initial EKG from this case there are voltage spikes of varying amplitudes, a chaotically wandering baseline, and a lead-specific artifact distribution. Other etiologies may mimic lead-connection artifact, but are readily distinguishable once they become familiar to the clinician.
60Hz AC interference should demonstrate almost exactly 60 deflections per second; the baseline typically will not wander and the amplitude will be constant or demonstrate orderly undulation. (Image retrieved from “Doktorekg.com,” http://www.metealpaslan.com/ecg/artef3en.htm)
Shivering artifact may or may not be accompanied by hypothermic ECG stigmata such as bradycardia or Osborne waves; note that the artifact is not confined to any single lead distribution. (Image retrieved from “LifeInTheFastLane.com,” http://lifeinthefastlane.com/ecg-library/basics/hypothermia/)
Artifact from resting tremor is typically of lower (physiologic) frequency and thus can mimic VT or a-flutter; relative to lead-connection artifact, tremor interference is pervasive, consistent, and of much longer cycle length.
The distinguishing hallmarks of lead-connection artifact are,
- It is confined to a specific lead distribution– the lead with inconsistent connectivity.
- There is a chaotically wandering baseline.
- The cycle lengths are short (30-70Hz ?) and grossly irregular.
- The amplitude is widely variable and randomly distributed.
When lead-connection artifact is recognized, operators can trouble-shoot the system for correctable problems. Often a “positional” solution can be temporarily utilized to acquire an acceptable tracing before the cables can be replaced. As in this case, when the origin of the artifact is unknown to the practitioner, it is not possible to investigate such a solution. The tragic coincidence presented here, where in the detection of STEMI was obscured by lead-connection artifact, illustrates that the potential significance of this issue.
While newer lead hardware has been made available, many operators continue to utilize the monitoring cables described in this case.
In this case, an anterior wall STEMI could not be appreciated due to artifactual interference. The patient was therefore transported to a non-PCI capable facility; subsequently, he did not receive definitive reperfusion until nearly five hours after his initial encounter with ACLS providers. The result was a catastrophic infarction from which he could not recover.
Operators should be familiar with the appearance of lead-connection artifact and maintain a high index of suspicion when checking and trouble-shooting this hardware.
- Equipment-related electrocardiographic artifacts: causes, characteristics, consequences, and correction. Patel SI, Souter MJ. Anesthesiology. 2008 Jan;108(1):138-48.
- Electrocardiographic artifacts during electroconvulsive therapy. Patel SI. J Electrocardiol. 2009 Jul-Aug;42(4):307-9. Epub 2009 Apr 2.
- Differential electrocardiographic artifact from implanted thalamic stimulator. Khan IA. Int J Cardiol. 2004 Aug;96(2):285-6. PMID: 15262047
- Electrocardiographic artifact caused by extracorporeal roller pump. Kleinman B, Shah K, Belusko R, Blakeman B. J Clin Monit. 1990 Jul;6(3):258-9. PMID: 2380757
- Electrocardiogram artifacts caused by deep brain stimulation. Constantoyannis C, Heilbron B, Honey CR. Can J Neurol Sci. 2004 Aug;31(3):343-6.
- ECG artifact produced by crystalloid administration through blood/fluid warming sets. Paulsen AW, Pritchard DG. Anesthesiology. 1988 Nov;69(5):803-4. PMID: 3189938 Free full text
- Life-threatening ECG artifact during extracorporeal shock wave lithotripsy. Schiller EC, Heerdt P, Roberts J. Anesthesiology. 1988 Mar;68(3):477-8. PMID: 3345012 Free full text
- ECG artifact due to deep brain stimulation. Martin WA, Camenzind E, Burkhard PR. Lancet. 2003 Apr 26;361(9367):1431. PMID: 12727397
- Electrocardiographic artifact induced by an electrical stimulator implanted for management of neurogenic bladder. Madias JE. J Electrocardiol. 2008 Sep-Oct;41(5):401-3. Epub 2008 Apr 28.
- An unusual electrocardiogram artifact: what is its source? [Gastric PM] J Reddy NK, Merla R, Pehlivanov ND, Pasricha PJ, Ware DL, Birnbaum Y. Electrocardiol. 2005 Oct;38(4):337-9. PMID: 16216608
- Unusual ECG artifact. [Infusion Pump] Graham MM. J Nucl Med. 1981 Jul;22(7):660. PMID: 7252570 Free full text
- Electromechanical association: a subtle electrocardiogram artifact.[Radial arterial impulse] Aslanger E, Yalin K. J Electrocardiol. 2012 Jan-Feb;45(1):15-7. Epub 2011 Feb 24.
- Tremor-induced ECG artifact mimicking ventricular tachycardia. Srikureja W, Darbar D, Reeder GS. Circulation. 2000 Sep 12;102(11):1337-8.
- Tremor-related artefact mimicking ventricular tachycardia. Ortega-Carnicer J. Resuscitation. 2005 Jun;65(3):243-4. PMID: 15919558
- Parkinson’s tremor mimicking ventricular tachycardia. Bhatia L, Turner DR. Age Ageing. 2005 Jul;34(4):410-1. PMID: 15955765 [Free full text]
- Pseudo-ventricular tachycardia: electrocardiographic artefact mimicking non-sustained polymorphic ventricular tachycardia in a patient evaluated for syncope. A Vereckei. Heart. 2004 January; 90(1): 81. PMCID: PMC1768000. [Free Full Text]
- Ventricular pseudo-bigeminy due to sustained myoclonus. Chung DK, Reed JR, Chung EK. Heart Lung. 1976 Nov-Dec;5(6):961-3. PMID: 1049219
- An unusual case of misdiagnosed ventricular tachycardia. Boos CJ, Khan MY, Thorne S. Emerg Med J. 2008 Mar;25(3):173-4.
- Pseudo ventricular tachycardia: a case report. Riaz A, Gardezi SK, O’Reilly M. Ir J Med Sci. 2010 Jun;179(2):295-6. Epub 2009 Aug 7. PMID: 19662493
- Tremor-induced ECG artifact mimicking ventricular tachycardia. Srikureja W, Darbar D, Reeder GS. Circulation. 2000 Sep 12;102(11):1337-8. PMID: 10982552 Free full text
- Tremor-induced ECG artifact mimicking ventricular tachycardia. Freedman B. Circulation. 2001 May 29;103(21):E112-2. PMID: 11382744 Free full text
- ECG artifact simulating supraventricular tachycardia during automated percutaneous lumbar discectomy. Lampert BA, Sundstrom FD. Anesth Analg. 1988 Nov;67(11):1096-8. PMID: 3189899
- Pseudo-atrial flutter/fibrillation in Parkinson’s disease. Prabhavathi B, Ravindranath KS, Moorthy N, Manjunath CN. Indian Heart J. 2009 May-Jun;61(3):296-7.
- The diagnostic use of respiratory artifact. Littmann L. J Electrocardiol. 2010 May-Jun;43(3):264-9. Epub 2009 Dec 2. PMID: 20399349
- Capsular contracture simulating myocardial infarction on ECG. Peters W, McEwan P. Plast Reconstr Surg. 1993 Mar;91(3):529-32. PMID: 8438025
- Influence of electrode misplacement on the electrocardiographic signs of inferior myocardial ischemia. Rudiger A, Schöb L, Follath F. Am J Emerg Med. 2003 Nov;21(7):574-7.
- [False diagnosis of myocardial infarction due to inversion of the electrocardiographic leads in the right limbs (author's transl)]. [Article in Spanish] Guijarro Morales A, Martos Ferrés F, Pagola Vilardebó C, Martín Jiménez V, Martí García JL, Peláez Redondo J. Med Clin (Barc). 1980 May 25;74(10):395-8.
- Delayed defibrillation caused by unexpected ECG artifact.[Bad lead selection and artifact] Stewart JA. Ann Emerg Med. 2008 Nov;52(5):515-8. Epub 2008 Apr 3. PMID: 18387704
- An unusual ECG artifact–results of a faulty recorder. Agarwal SK. JAMA. 1979 Aug 17;242(7):617-8. PMID: 448997
- Electrocardiographic ST-segment depression: confirm, deny, or artifact? [Filters] Wong DH. Anesthesiology. 2008 Aug;109(2):352; author reply 352. PMID: 18648245 Free full text
- False ST elevation in a modified 12-lead surface electrocardiogram. Toosi MS, Sochanski MT. J Electrocardiol. 2008 May-Jun;41(3):197-201. Epub 2008 Mar 14. PMID: 18342880
- Myocardial infarction or technical artifact? [Electrode misplacement] MacKenzie R. J Insur Med. 2006;38(4):289-92. PMID: 17323759
- Simulation of anteroseptal myocardial infarction by electrocardiographic filters. Burri H, Sunthorn H, Shah D. J Electrocardiol. 2006 Jul;39(3):253-8. Epub 2006 Feb 28. PMID: 16777511
- Electrocardiographic artifact mimicking acute myocardial infarction. Siddiqui MA, Munugoti S, Khan IA. Int J Cardiol. 2003 Jan;87(1):99-101. PMID: 12468060
- Artifactual electrocardiographic change mimicking clinical abnormality on the ECG. Chase C, Brady WJ. Am J Emerg Med. 2000 May;18(3):312-6. PMID: 10830688
- An unusual electrocardiogram artifact in a patient with near syncope. Aslanger E. J Electrocardiol. 2010 Nov-Dec;43(6):686-8. Epub 2010 Jun 2. PMID: 20553822
- [Brugada or not-Brugada: misdiagnosis of recorder-induced artifact]. [Article in German] Z Kardiol. 2002 Dec;91(12):1061-3. Martius P, Krämer H.
- ECG artifacts and heart period variability: don’t miss a beat! Berntson GG, Stowell JR. Psychophysiology. 1998 Jan;35(1):127-32. PMID: 9499713
- Pacemaker malfunction: fact or artifact? Murdock DK, Moran JF, Stafford M, King L, Loeb HS, Scanlon PJ. Heart Lung. 1986 Mar;15(2):150-4. PMID: 3633245
- An example of apparently normal electrocardiogram originating from incorrect electrocardiographic acquisition in a patient with ST-segment elevation myocardial infarction. J Electrocardiol. 2010 May-Jun;43(3):222-3. Epub 2010 Mar 23. Aslanger E, Yalin K, Golcuk E, Oncul A.
- Double counting of heart rate by interpretation software: a new electrocardiographic sign of severe hyperkalemia. Am J Emerg Med. 2007 Jun;25(5):584-6. Littmann L, Brearley WD Jr, Taylor L 3rd, Monroe MH. PMID: 17543665
- Classic EKG changes of hypothermia. Mareedu RK, Grandhe NP, Gangineni S, Quinn DL. Clin Med Res. 2008 Dec;6(3-4):107-8. PMID: 19325173
- Common electrocardiographic artifacts mimicking arrhythmias in ambulatory monitoring. Márquez MF, Colín L, Guevara M, Iturralde P, Hermosillo AG. Am Heart J. 2002 Aug;144(2):187-97. PMID: 12177632
- Hiccup as an electrocardiographic artifact simulating arrhythmias. Am Heart J. 2003 Oct;146(4):E15; author reply E16. Cheng TO. PMID: 14564338
- “The Bait and Switch”. [Limb lead reversal due to inverted lead coupling conceals STEMI] Tom Bouthillet. EMS 12-Lead. 01-22-2011. Retrieved from: http://ems12lead.com/2011/01/the-bait-and-switch/
- “Unusual EKG/ECG Pattern: You don’t see this everyday”. The Happy Hospitalist. 04-15-2011. Retrieved from: http://thehappyhospitalist.blogspot.com/2011/04/unusual-ekgecg-pattern-you-dont-see.html